Laser Point Laser And Energy Meters
Laser Point manufactures hi tech laser power and energy meters. It is the choice of accuracy in laser measurement. Wavelength OptoElectronic is the partner of the Laser Point brand of laser and energy meters for Southeast Asia.

Products

Descriptions

Request for Quotation

Thermal Sensors

Sensor Meters

OEM Solutions
Part Number  Power Range  Max Intermittent Power  Energy Range  Useful Aperture  Spectral Range  Absorber  Cooling 
A02D12BBF  0.1 mW  200 mW  200 mW  1 mJ  200 mJ  10 mm  0.19  25 µm  BBF  Convection 
A2D12BBF  1 mW  2 W  2 W  1 mJ  2 J  10 mm  0.19  11 µm  BBF  Convection 
A2D12HPB  1 mW  2 W  2 W  1 mJ  2 J  10 mm  0.19  25 µm  HPB  Convection 
A5D12BBF  1 mW  5 W  7.5 W  10 mJ  5 J  10 mm  0.19  25 µm  BBF  Convection 
A10D12HPB  10 mW  10 W  15 W  10 mJ  15 J  12 mm  0.19  11 µm  HPB  Convection 
A10D20BBF  10 mW  10 W  15 W  10 mJ  15 J  20 mm  0.19  25 µm  BBF  Convection 
A10D20HPB  10 mW  10 W  15 W  10 mJ  15 J  20 mm  0.19  11 µm  HPB  Convection 
A30D25HPB  20 mW  30 W  45 W  50 mJ  45 J  25 mm  0.19  11 µm  HPB  Convection 
A40D25HPB  20 mW  40 W  60 W  50 mJ  60 J  25 mm  0.19  11 µm  HPB  Convection 
A40D40HPB  20 mW  40 W  60 W  50 mJ  60 J  40 mm  0.19  11 µm  HPB  Convection 
A40/200D25HPB  150 mW  40 W  200 W  200 mJ  200 J  25 mm  0.19  11 µm  HPB  Convection 
A40/200D40HPB  100 mW  40 W  200 W  150 mJ  200 J  40 mm  0.19  11 µm  HPB  Convection 
A40/200D60HPB  200 mW  40 W  200 W  250 mJ  200 J  60 mm  0.19  11 µm  HPB  Convection 
A200D25HPB  0.2 W  200 W  250 W  0.5 J  250 J  25 mm  0.19  11 µm  HPB  Forced Air 
A200D25SHC  0.2 W  200 W  250 W  0.5 J  250 J  25 mm  0.19  11 µm  SHC  Forced Air 
A200D40HPB  0.2 W  200 W  250 W  0.5 J  250 J  40 mm  0.19  11 µm  HPB  Forced Air 
A200D40SHC  0.2 W  200 W  250 W  0.5 J  250 J  40 mm  0.19  11 µm  SHC  Forced Air 
A200D60HPB  0.3 W  200 W  250 W  1 J  250 J  60 mm  0.19  11 µm  HPB  Forced Air 
A200D60SHC  0.3 W  200 W  250 W  1 J  250 J  60 mm  0.19  11 µm  SHC  Forced Air 
A300D60HPB  0.5 W  300 W  400 W  1 J  400 J  60 mm  0.19  11 µm  HPB  Forced Air 
A600D40HPB  0.5 W  600 W  800 W  1 J  800 J  40 mm  0.19  11 µm  HPB  Forced Air 
A600D60SHC  0.5 W  600 W  800 W  1 J  800 J  60 mm  0.19  11 µm  SHC  Forced Air 
W200D40HPB  0.2 W  200 W  300 W  1 J  300 J  40 mm  0.19  11 µm  HPB  Water 
W200D40SHC  0.2 W  200 W  300 W  1 J  300 J  40 mm  0.19  11 µm  SHC  Water 
W600D30HPB  0.5 W  600 W  800 W  2 J  800 J  30 mm  0.19  11 µm  HPB  Water 
W600D30SHC  0.5 W  600 W  800 W  2 J  800 J  30 mm  0.19  11 µm  SHC  Water 
W600D70HPB  0.5 W  600 W  800 W  2 J  800 J  70 mm  0.19  11 µm  HPB  Water 
W600D70SHC  0.5 W  600 W  800 W  2 J  800 J  70 mm  0.19  11 µm  SHC  Water 
A1200D60SHC  2 W  1200 W  tbd W  5 J  1200 J  60 mm  0.19  11 µm  SHC  Forced Air 
W1500D40HPB  4 W  1500 W  2250 W  5 J  2250 J  40 mm  0.19  11 µm  HPB  Water 
W1500D40SHC  4 W  1500 W  2250 W  5 J  2250 J  40 mm  0.19  11 µm  SHC  Water 
W3000D55HPB  6 W  3 kW  4.5 kW  n.a.  n.a.  55 mm  0.19  11 µm  HPB  Water 
W3000D55SHC  6 W  3 kW  4.5 kW  n.a.  n.a.  55 mm  0.19  11 µm  SHC  Water 
W6000D55SHC  15 W  6 kW  9 kW  n.a.  n.a.  55 mm  0.19  11 µm  SHC  Water 
10BBD12L  2 mW  10 W  15 W  50 mJ  10 J  12 mm  0.4  2 µm  BB  Convection 
10BBD25  2 mW  10 W  15 W  50 mJ  10 J  25 mm  0.4  5.2 µm  BB  Convection 
10UVAD25  2 mW  10 W  15 W  50 mJ  10 J  25 mm  0.25  0.4 µm  UVA  Convection 
10UVCD25  2 mW  10 W  15 W  50 mJ  10 J  25 mm  0.19  0.25 µm  UVC  Convection 
20BBD40  40 mW  20 W  30 W  100 mJ  20 J  40 mm  0.4  5.2 µm  BB  Convection 
20UVAD40  40 mW  20 W  30 W  100 mJ  20 J  40 mm  0.25  0.4 µm  UVA  Convection 
20UVCD40  40 mW  20 W  30 W  100 mJ  20 J  40 mm  0.19  0.25 µm  UVC  Convection 
A10D12DIF  10 mW  10 W  15 W  10 mJ  15 J  12 mm  0.2  2.1 µm  DIF  Convection 
A30D12SHCL  20 mW  30 W  45 W  100 mJ  45 J  12 mm  0.2  1.1 µm  SHC  Convection 
A30D18DIF  25 mW  30 W  45 W  100 mJ  45 J  18 mm  0.2  2.1 µm  DIF  Convection 
A40D33DIF  25 mW  40 W  60 W  100 mJ  60 J  33 mm  0.2  2.1 µm  DIF  Convection 
OEM SOLUTION SERIES  DESCRIPTION 
SD  Sensor Disk  Thermal Sensor Disks supplied with bare wires, no housing, no cooling system and no electronics. SD sales opportunity is evaluated case by case by Laser Point after direct discussion with end user customer. 
CS  Cased Sensor  Thermal Sensor Disks supplied with housing (CSA Air, CSW  Water) 
AH  Amplified Head  Cased Thermal Sensor Disks supplied with electronics for analog signal amplification, filtering and acceleration. 
BC  Blink Fast Response Cell  Blink Fast Response Cell supplied with bare wires, no housing, no cooling system and no electronics. BC sales opportunity is evaluated case by case by Laser Point after direct discussion with end user customer. 
BO  Blink OEM  Blink Cell supplied with housing, optimized heat exchanger, electronics for signal amplification, filtering and acceleration. 
BA  Blink Amplified  Blink Cell supplied with housing, optimized heat exchanger, electronics for analog signal amplification, filtering and acceleration. 
FIT H  Fast Integrated Thermopile  Power Probe supplied with housing and optimized heat exchanger 
 Thermal Sensors
Thermal methods of measuring power and energy are those in which radiant energy is absorbed and converted into heat, which generates a temperature rise in the absorber. The absorbed energy is then measured through a function that takes into account the temperature gradient between the hot area (where the laser strikes) and a cool area (where the generated heat is dissipated). This measurement can be done by means of thermocouple arrays (thermopile). The temperature difference will generate a voltage at the end of each single thermocouple and the resulting total voltage will be proportional to the incident power or energy.
 Sensor Meters
Different sensor meters are available, allowing one to read measurements on standalone portable instruments, on a PC screen or simply storing them on a memory device.
 OEM Solutions
Laser Point has developed a series of compact power sensors that can be easily integrated into laser machines to monitor the power during the operation. For more than two decades Laser Point has been supplying OEM power & energy sensors to some of the world’s leading laser manufacturers. R&D activities are still strong, especially on new applications or on the design of new detectors.
For more products and information, click here.